Elastic Scattering Angle (& diffraction angle) of Electrons  Practical Electron Microscopy and Database   An Online Book  

Microanalysis  EM Book https://www.globalsino.com/EM/  
Rutherford crosssection gives convenient formulas to calculate the elastic scattering angle and total elastic cross section so that it is often used to describe the elastic scattering of electron. However, its accuracy becomes worse in lowenergy region or in high atomicnumber materials. As atomic number (z) increases the elastic scattering cross section becomes larger and thus the possibility of scattering into large angles becomes higher which increases the backscattering coefficient. In general, the inelastic scattering of electrons occurs at smaller angles than that for elastic scattering. Figure 4754 shows angular distributions of various scattered electrons as a function of scattering angle in EMs.
Figure 4754. (a) Elastic scattering of electrons with crystalline and amorphous specimens, (b) Phonon scattering, (c) Innershell ionization losses, and (d) Plasmon losses. For crystals, for instance, the wavelength of 200kV TEM is 0.00251 nm, and d_{111} for Si is 0.3135 nm, thus the angle between direct beam and Si 111 spot is given by (see page3580),  [4754] 

